Payment & Shipping Terms:
|Application:||Outdoor Lightings, Household Appliance||IP Grade:||IPX3 And IPX4|
|Warranty:||12 Months||Control System:||PLC+ Touch Screen|
environmental test chamber,
temperature humidity test chamber
IP03 / IP04 Rainproof Environment Test Equipment R200 to R1600 Oscillating Tube Selectable
|1. Commodity||IP03 / IP04 Rainproof Environment Test Equipment|
Application Background: The water from the nature (rainwater, seawater and river water and etc.) would cause damage to products and material. The damage includes corrosion, color fading, deform, strength degradation, mildew and so on. Especially, it will make the short circuit to electronic products. So waterproof testing is a must test for some special products and materials.
Application; outdoor lighting, household appliance, auto spare parts and other electronics products.
|Function: This machine is mainly applied to simulate the equipment (like electronic products, lighting, electronic components, automobile, motorbike and its spare parts) under the raining conditions, and to check if its physical and other relative performances can reach the requirements.|
Meet the requirements of IEC60529:1989+A1:1999+A2:2013 clauses 14.2.3, 14.2.4 and figure 4, GB7000.1 clauses 9.2.4, 9.2.5 and figure 9.4.
● Apply to IPX3 and IPX4 testing, R1/R2/S1/S2 testing
● Structure: adopts double pillars structure, easy to install and smooth running.
● The shells and turntable are made by SUS304# stainless steel, to make sure rustless for long time use.
● Built-in water tank, with ball valve to control the inflow, and with low water-level alarm to prevent from damage caused by hydropenia. The thermometer is to display the temperature of the tank.
● Oscillating tube drive: adopts step motors, made by PANASONIC (with features of low noise and high accuracy).
● Turntable drive: adopts Chinese superior step motors, the test speed and angle can be set up by touch screen, the speed is within the range of standard can not be adjustable, and can control the positive and negative direction (for live test, to prevent the winding) .
● Test time settable: 0~999min (adjustable), on-off time can be control by program: off (0~999min)-on (0~999min)-off (0~999min, It is apply to live test.
● Touch screen control:
A: IPX3 test: as per IEC60529 regarding IPX3 test.
B: IPX4 test: as per IEC60529 regarding IPX3 test.
C: User-defined: as per user’s demands, test time, angle, oscillating tube speed, and turntable speed can be freely to set up.
● Control system adopts 7 inch touch screen and PLC( Panasonic), the electrical components are made by LG, OMRON and other famous brands.
● Filter: equip with high accuracy filter, can filter out impurities, and the filter element can be changeable.
● Equip with necessary accessories, like filter element, pin, hex screw and etc.
|6. Specification||Structure Parameter|
|Oscillating tube||R1200mm, R1000mm, R800mm, R600mm, R400mm, R200mm|
|Outer dimension||L 3200*D 2400*H 2400mm|
|Dia. of blowhole||Dia. 0.4mm|
|Inner dia. of oscillating tube||15mm|
|Diameter of tuntable||Dia. 800mm|
|Height of turntable||1350-1950mm|
|Turntable Speed||1-5r/min (adjustable)|
|Rotation mode||Positive/positive and negative alternate|
|Oscillate angle||120º (IPX3), 350º (IPX4) or user-defined|
|Oscillate speed||Back and forth at 1200: 4s; back and forth at 3600:12s or user-defined|
|Power supply for equipment||220V|
|Power supply for EUT||220V|
|Protection device||Leakage protection, Lack of water protection, short-circuit protection|
|Case||304# stainless steel||1 suit|
|Water tank||304# stainless steel, 50L||1 piece|
|Turntable||304# stainless steel||1 piece|
|Oscillating tube||304# stainless steel||1 piece|
|Needle||Φ0.4mm stainless steel||100 pieces|
|Needle base||M5-6||20 pieces|
|Water pump||Stainless steel||1 piece|
|Flowmeter||Glass panel, adjustable||1 piece|
|Pressure gauge||Anti-seismic stainless steel/Y-60BFZ||1 piece|
|Ball cock||ND15||1 piece|
|Filter||10 inch filter||1 piece|
|Touch screen||7 inch color screen||1 piece|
|PLC controller||FR-XO L30R (PANASONIC)||1 piece|
|Step motor for tube||PANASONIC||1 piece|
|Step motor for turntable||57#||1 piece|
|L-voltage appliance||LG||1 batch|
|Power Switch||OMRAN||1 piece|
2. Calibration Certificate
4. filter element
5. needle: 100 pieces
6. Needle base; 20 pieces
7. three-core cable: 3m
|9. Working Condition||User shall provide the below conditions and make the connection|
● Flat floor, Smooth drainage, the floor and wall with water repellent treatment, set the gutter and drainage ditch
● The ground bearing shall not less than 200KG/m2
● With excellent ventilation
● Nonviolent vibration
● Non-strong electromagnetic field
● No flammable, no explosive and no corrosivity substance, no dust
● With appropriate use and maintenance space
● Rear side, left side, right side, top: 1400mm
● Front side: 1000mm
Air Pressure: 86kPa～106kPa
|10. Power Condition|
|Voltage, frequency||AC380V±10% , 50Hz±10%, 3 phase 4 wires+earth wire|
● Request to equip with appropriate air switch for the equipment at the site, and the air switch is only work for this equipment
● Forbidden to use knife switch and power plug
● No more than 2m between air switch and equipment
● Request to equip with water supply, and hydraulic pressure shall not lower than 0.2Mpa, and install 4’’ faucet.
● The faucet should be installed at the position away from the equipment no more than 2m.
2. Test Method Instruction
2.1 IPX3 and IPX4 test method:
|hole count||Total flow||hole count||Total flow|
● Put the sample on the turntable
● Preset the water flow: Choose the suitable oscillating tube according to the sample size. And adjust the height of turntable to make the distance from top of sample to oscillating tube no exceed 200mm.
● Apertures out of the angle of 600 should be block to make the water no raining on the sample.
● Power on the turntable to make it rotates, power on the sample to make it works.
● Set up the angle, speed, test time and flow at the control panel. Then press start button, the water pump will auto open,system runs and record the test time, oscillating tube runs at the angle of 1200 with speed of 4S back and forth. Then check the flowmeter, if it can not meet the requiremnet, make some adjustment. When time’s up, the test will auto stop, then cut off the power, shut down the inlet, and take down the sample.
Operation steps are basically as same as the IPX3 test, the difference as below:
● Apertures on the oscillating tube shall not be block.
● Swing angle is approach to 3500
● The circle shall be 2*3500 with speed of 12S or undefined.
4. Third-Lab Calibration Certificate
Contact Person: sales