Payment & Shipping Terms:
|Standard:||IEC60065, IEC60950-1, IEC60884-1||Force Arm:||0-200mm|
|Test Function:||For Testing The Mechanical Strength Of Direct Plug-in Equipment||Weights:||2.5N (1 Set), 1N (1 Set)|
|Accessories:||All-purpose Plug, UL Plug, AS/NZS Plug||Material:||Stainless Steel|
electrical plug tester,
torque testing equipment
Plug Torque Test Apparatus for Devices Forming a Part of the Main Plug IEC60065 IEC60884-1
Plug torque test device is designed according to IEC60065 clause 15.4.1 and figure 11, IEC60950-1 clause 4.3.6, IEC60884-1 clause 14.23.2 and etc..It is applied to check the socket outlet mechanical strength of direct plug-in equipment. A device provided with pins intended to be introduced into fixed socket-outlets shall not impose undue strain on these socket-outlets, the torque shall not exceed 0.25Nm.
|Distance between contact surface and axis surface||8mm|
|Weights||2.5N (1 set), 1N (1 set)|
|Configuration||all-purpose plug (1 set, apply to GB and BS plugs),UL plug (1 set) , AS/NZS plug (1 set)|
|Note: other plugs can be customized|
3. working principle:
Compliance is checked by engaging the device, as during intened use, with the socket-outlet of a test apparatus. The balance arm of the test apparatus pivots about a horizontal axis through the centre lines of the contact tubes of the socket-outlet at a distance of 8mm behind the engagement face of the socket-outlet.
With the device not in engagement, the balancing arm is in equilibrium, the engagement face of the socket-outlet being in the vertical position.
After the device has been engaged, the torque to be applied to the socket-outlet to maintain its engagement face in the vertical plane is determined by the position of weight on the balance arm, the torque shall not exceed 0.25Nm.
Contact Person: sales