Payment & Shipping Terms:
|Standard:||IEC60529, IEC60598||Material:||Brass, SUS304 Stainless Steel|
|Warranty:||12 Months||Shipment:||By Sea|
|Application:||LED Luminaires||Name:||Spray Nozzle Test Equipment|
IPX4 Spray Nozzle Tester,
SUS304 Spray Nozzle Tester,
150L Environmental Test Chamber
IEC60529 IPX3 and IPX4 Spray Nozzle Test Equipment for Outdoor LED Luminaires
IPX3/4 spray nozzle test equipment can meet the requirement of IEC60529 clauses 14.2.3 and 14.2.4 and figure 5, it is it is apply to verify the protection against spraying water (IPX3) and splashing water (IPX4) for large size sample. The equipment is consisted by spray nozzle, water tank and control box. This equipment is to simulate the raining conditions to check the physical performance of outdoor luminaires, household appliance, auto parts and etc.
● For IPX3 and IPX4 tests.
● The spray nozzle is processed by CNC technology, the head is made of pure brass, and the surface is mirror-polished. The moving shield is made of aluminum, and the other parts are made of SUS304 stainless steel.
● The surface of the nozzle head is sprayed with an anti-rust coating, which can effectively prevent the watermark on the surface of the nozzle from producing patina and affecting the appearance.
● The spray nozzle has 121 holes with diameter 0.5mm (1 hole at center, 1 inner circles of 12 holes at 300 pitch, 4 outer circles of 24 holes at 150pitch), completely meet the requirement of IEC60529.
● The control box includes timer, power switch button, emergency stop button, start button and flowmeter.
|Outer size (L*W*H)||
870mm*600mm*950mm (control box)
370mm*115mm*165mm (spray nozzle)
|Spray nozzle diameter||102mm|
|Number of holes||121|
1 hole at center
1 inner circles of 12 holes at 300 pitch
4 outer circles of 24 holes at 150pitch
|Outer diameter of water inlet||Φ16mm|
|Outer diameter of water pipe||Φ28mm|
|Flowmeter adjustment method||Manual adjustment|
|Water pressure range||50-150Kpa|
|Flowmeter display||Glass flow meter|
|Capacity of water tank||150L|
1). The user is required to configure a water source for the equipment at the installation site, with a water pressure not less than 0.2Mpa, and install a 4/8 inch faucet.
2). The faucet is not more than 2 meters away from the equipment
|Requirement for the site||The ground is level, the drainage is unobstructed, the ground and walls are waterproofed, and the floor drain or drainage ditch is set|
Contact Person: sales