Payment & Shipping Terms:
|Standard:||IEC60065 Figure 11, IRAM 4229||Application:||Dricet Plug-in Equipment|
electrical plug tester,
torque testing equipment
Torque Tester Apparatus for Direct Plug-In Equipment with Different Adaptors According to IEC60065
The torque tester apparatus is designed according to IEC60065, it is a test to check various elements that are connected to sockets. A device provided with pins intended to be introduced into fixed socket-outlets shall not impose undue strain on these socket-outlets.
Compliance is checked by engaging the device, as during intended use, with the socket-outlet of a test apparatus as shown in figure 11. The balancing arm of the test apparatus pivots about a horizontal axis through the centre lines of the contact tubes of the socket-outlet at a distance of 8 mm behind the engagement face of the socket-outlet.
With the device not in engagement, the balancing arm is in equilibrium, the engagement face of the socket-outlet being in the vertical position.
After the device has been engaged, the torque to be applied to the socket-outlet to maintain its engagement face in the vertical plane is determined by the position of a weight on the
The torque shall not exceed 0,25 Nm.
|Distance||8 mm behind the engagement face of the socket-outlet|
all-purpose plug for GB and BS plugs
|According to||IEC 60065, IRAM 4229|
IEC60065 Figure 11
Contact Person: Ms. Penny Peng