Payment & Shipping Terms:
|Standard Compliant:||IEC60335-2-15 Clause 15.2 And Figure 101||Test Solution:||30ML With 1% NACL|
|Application:||For Testing The Connectors With Spillage Prtoection||Inclined Angle:||20°|
Connectors Spillage Test Apparatus,
30ML Spillage Test Apparatus,
Spillage Prtoection Spillage Test Equipment
IEC60335-2-15 30ML Spillage Test Apparatus for Testing the Connectors with Spillage Prtoection
The spillage test apparatus meets the requirements of IEC60335-2-15 clause 15.2 and figure 101. It’s used to check the connectors with spillage protection to ensure that the connectors are not affected by liquids when installed in a rated voltage base according to the manufacturer’s installation instructions.
the spillage test is carried out with the appliance deviating from the normal position of use by an angle not exceeding 5°. The kettles are placed on a plane inclined at an angle of 20° to the horizontal and 30ml of water containing approximately 1% NACL is poured onto the top rim in the most unfavourable place. The solution is poured steadily through a tube having an inner diameter of 8mm over a period of 2s, the lower end of the tube being 200mm above the appliance. A further quantity, equal to 15% of the rated capacity of the kettel, is then added as quickly as possible.
|Tube dimension||Inner diameter: Ø8mm, length 3mm|
|Distance between the lower end of the tube and specimen||200mm|
|Water injection method||Double water injection pipes, one for tap water, and the other for salt water|
|Flow control||Manual control|
|Lifting method||Manual screw adjustment of the lifting table|
Contact Person: sales