Payment & Shipping Terms:
|Product Name:||Lamp Cap Torque Test Meter||Test Fixture:||E26/E27, E14, B22d, G5/G13, E40 (optional）|
|Measuring Range:||0-10N.m||Configuration:||Secondary Meter, Torque Device And Test Fixtures|
|Operation System:||5 Inch Touch Screen||Working Power:||220V±10%, 50Hz/60Hz|
B22d Lampcaps Digital Torque Meter,
5 Inch Touch Screen Led Testing Equipment,
E40 Lampcaps Digital Torque Meter
Lampcaps Digital Torque Meter 5 Inch Touch Screen For E26/E27 E14 B22d G5/G13 E40
1. Introduction of Lampcap Torque Meter
Lamp Torque Meter is a tool to detect the torque of different lamp caps such as energy-saving lamps, incandescent lamps, high-pressure mercury lamps, and sodium lamps.
HP-10 digital torque meter is a tool for testing the mechanical strength of lamp caps. It can be equipped with different fixtures to measure the torque value of lamp caps of different lamps such as LED bulbs, HID lamps and CFL lamps. The equipment consists of secondary meter, torque device and test fixture (for lamp holders E27/E26, E14/E12, B22d, G5/G13). The peak and lower limit alarms can be preset through the touch screen, and the torque value can be displayed in real time.
2. Specification of Digital Torque Meter
|Control unit||5-inch LCD touch screen|
|Fixed mode||Fix the lampcap with the fixture, and rotate the lampcap forward and reverse to measure the torque value of the lamp cap|
|Fixture||E26/E27, E14, B22d, G5/G13 and E40 (optional)|
|Temperature Compensation Range||-10℃~60℃|
|Working condition||-0~40℃, ≤75% R.H.|
|Storage condition||-20℃~50℃, ≤75% R.H.|
|Working Power||220V±10%, 50Hz/60Hz (single phase)|
3. Working Principle of lampcap torque meter
The instrument is mainly composed of two parts: torque device and secondary instrument. The key to torque testing accuracy is the torque sensor. The instrument adopts strain-type torque sensor, which has the advantages of high reliability, high precision and long life. A resistance strain gauge is attached to the elastomer of the sensor to form a Wheatstone bridge, and an excitation voltage is applied across the bridge. Under the action of torque, the elastic body is deformed, and the resistance of the strain gauge changes, which makes the bridge unbalanced, and the bridge outputs a voltage signal that is linearly related to the torque. The secondary instrument amplifies the voltage signal and sends it to the microprocessor for data processing and display after A/D conversion
1). Can you provide third-lab calibration certificate?
Answer: Yes, the equipment can send to the third-labs authorized by ISO17025 and get the calibration certificate, but the buyer should undertake the charges
2). How can we order this item?
Answer: please send inquiry to us, our salesman will contact you within 24 hours.
Contact Person: Ms. Penny Peng